Our 33 Litre Front Loading Benchtop Autoclave, available in Classic and Autofill versions, is ideal for laboratory research and is low in water. Have a question about Autoclaves, Sterilizers or Steam Generators? Our FAQ page has the answers. All the machines are supplied with a user manual but sometimes it is not consulted! Astell offer training to operators, and this is a service I recommend to ensure.
|Published (Last):||10 January 2014|
|PDF File Size:||14.52 Mb|
|ePub File Size:||18.85 Mb|
|Price:||Free* [*Free Regsitration Required]|
Astell can custom build the Benchtop range to your specifications.
Load the chamber and close the door. A choice of 33, 43 and 63 litre models Self contained aastell with no installation required – just fill it with water and plug it in and the machine is ready to be used No mains water supply or drain necessary Fully programmable 5. Please contact Astell for further details. Press the function key under the LCD display.
Frequently Asked Questions | Astell UK
Where a Load Sensed Timing option- sales part. Do you supply autoclave spare parts separately? Astell Scientific offer servicing contract options for all the sterilization equipment we manufacture. A temperature threshold is derived automatically within each Program exactly 2 Deg. There are up to 8 Programs with a number of parameters which may be set.
The Letters correspond with the configuration items that are set ON. This system is integral with and dependent upon the use of the Load Sensing system. Do our autoclaves need manuall be connected to autoclaave vent system? Press it again and Recorder power will turn off, and so on.
If not used for liquids at allit may be set to 99C.
Autoclave service and maintenance – advice from the experts
The mnaual may be set using the Clock Setting section in the Supervisor “Options” An illustration of Paper Loading is at the back of this manual. When the load reaches 1.
Prevents starting a cycle with a dangerously hot chamber. Benchtop autoclaves brochure KB. If the Printer is fitted then the Printout Heading and cycle description will be printed. Because of the increased temperature differential between load and chamber, the wstell heatsup much faster over the last few degrees.
This setting will be as required to ensure correct operation. Most keypresses are signalled by a “Beep” tone.
Astell MXN451 5000 – 6000 Series Autoclave programming instructions
Heating to C Astell creates a bespoke all-stainless-steel autoclave for toric contact lens producer PolyDev. The slow reduction reduces boiling over of media. This item not yet used. Aktoclave printer can also be astsll to provide reports of cycle settings and servicing information. By introducing steam between vacuum pulses, air is forced out of the autoclave chamber under pressure; the best method for removing air from porous type loads and discard waste.
In combination with the jacket option, a drying stage then begins to ensure loads are touch dry upon removal.
It is designed to prevent the chamber exceeding asteell Dry design temperature which might cause a hazard and provides some protection for some types of load.
This printout takes approximately seconds to complete.
All device changes and signatures are electronically logged and saved in a file format that cannot be altered in any way. For Example, this allows a load to be placed in the chamber in the afternoon, which will start automatically before start of work next day.
Here the Freesteam time was set to 8 Min. These 4 switches control functions that cannot be set by the software or Configuration. However, please note that in normal circumstances some water will remain in the chamber at awtell end of each cycle, so the water required to reach the minimum level will be relatively small.
You may now Start the Program Water Level Sensing The water level sensing system is designed to ensure sufficient water supply for a cycle and protects the heaters should excessive water loss occur.